To ensure accurate results and proper function of all parameters, the meter should be checked for accuracy daily.
- A full calibration of your Mesa Medical meter should be performed periodically (monthly is suggested but it is ultimately up to the user).
- Use the 4.0 and 10.0 pH buffers when performing a calibration on the pH circuit instead of only the 7.0 pH buffers. Using the 4.0 and 10.0 pH buffers ensure that the pH functions properly and will identify some issues that the 7.0 pH buffer alone may not be able to detect.
Let’s talk DialyGuard!
Talk with one of our experts to get your questions answered and see how we can help you solve your meter and solution pain points.